AM-Y200 ventilator is a new-type of pneumatic-driven and electronic-controlled ventilator which is designed to meet the special needs of clinical respiratory management and respiratory therapy for infants and children with respiratory disorders.
With the feature of innovative functions, convenient operation, less gas consumption, compact but reasonable structure, and outstanding safety and reliability, it is an indispensable instrument in emergency and ICU rooms, and an essential tool to rescue the life of infants and children.
-AR-Y200 ventilator can work with some innovative ventilation modes, e.g., constant flow
mode, pressure limit mode and time cycle ventilation mode, and also supports PEEP and IMV functions.
|Working Mode (Ventilation Mode)||Controlled Mechanical Ventilation (CMV),
Intermittent Mandatory Ventilation (IMV),
Continuous Positive Airway Pressure (CPAP),
Manual Ventilation (Manual).
|Inspiratory to Expiratory Ratio||1:1~1:3|
|Special Order I:E Range||10:1~1:9.9|
|Inspiratory Flow||3 L/min~20 L/min|
|Tidal Volume||Minimum 10 mL~300 mL|
|Minute Volume||Max. Ventilation volume greater than 5 L/min|
|Oxygen Concentration of Output Gas||21%~100%|
|Ventilator System Compliance||Not greater than 3×10-2 mL/Pa|
|Positive End-Expiratory Pressure (PEEP)||0kPa~2.0kPa|
|Inhalation Safety Valve||Safety valve shall be opened when the airway pressure is within -300 Pa ~ -800 Pa|
|Synchronous Trigger||Pressure Trigger||-1.0~0kPa (Special Order)|
|Flow Trigger||0~10L (Special Order)|
|Maximum Safe Pressure||Not greater than 10kPa|
|Continuous Positive Airway Pressure (CPAP)||0kPa~2.0kPa|
|Monitoring Tidal Volume||Real-time monitoring within the adjustment range of tidal volume: ＜100 mL: ±10 mL; ≥100 mL: ±15 %;|
|Monitoring Oxygen Concentration||21%~100%|
|Power Failure Alarm||Audible alarm sounds immediately in case of power outage and lasts at least for 120s|
|Gas Source Failure Alarm||Audible and visual alarm sounds within 6s after gas source interruption occurs and lasts at least for 120s|
|Airway Pressure Lower Limit Alarm Range||PEEP +0.2kPa|
|Airway Pressure Upper Limit Alarm Range||Inspiratory pressure +0.4 kPa|